FCEM marionette has been properly oriented looking at the back reflection from the open flange. We did the adjustment with open loop.
- TX has been adjusted moving the marionette TX motor (number 5)
- TY has been adjusted adding a DC on the coils (0.15 V)
Then we centered oplev PSD and closed the loop. We needed to further adjust the loop set-points to find the flashes. (TX = -100, TY= -500)
We have also realigned PSD2 (TZ), PSD3 (MIR_TY, MIR_TX) and PSD 4 (MIR_Z)
Config file has been updated with the new values.