Today in the EERoom we replaces the six ADC7674 VME boards (SN: 10, 18, 20, 38, 15, 21) used by INJ, ENV and PSL sub-systems. We replace these boards by DaqBoxes with mezzanines ADC2378.
DaqBox 122 | ADC +/-20V 183 | ADC +/-20V 184 | ADC +/-20V 197 | ADC +/-20V 244 |
DaqBox 123 | ADC +/-20V 186 | ADC +/-20V 194 | ADC +/-20V 199 | ADC +/-20V 206 |
DaqBox 124 | ADC +/-20V 145 | ADC +/-20V 148 | ADC +/-20V 171 | ADC +/-20V 196 |
We added three Ethernet cables connected to the EERoom switch for the DaqBox connection. All unused optic fibers and timing cables have been removed.
Metalic part of BNC to Lemo adaptators have been isolated with heat-shrinkable sleeves.
We took the oportunity to clean the wiring of the whole rack, that's why the task took more time than expected.
Due to cable length constraints, three channels have been moved from already installed DaqBoxes:
INJ_SIB1_PELTIER moved from | DaqBox 40 - ADC +/-10V 30 ch7 | to | DaqBox 123 ADC +/-20V 199 ch6 |
INJ_IMC_REFL_I_POST moved from | DaqBox 42 - ADC +/-10V 48 ch0 | to | DaqBox 124 ADC +/-20V 196 ch0 |
ENV_IB_CT_FINGER_ACC_Y moved from | ADC7674 10 ch10 +/-20V | to | DaqBox 42 - ADC +/-10V 48 ch0 |
Gain of these channels will be changed in the configuration file according their new destination.
We still have to modify configuration files and check that all channels are properly aquired in the DAQ.