R. Cavalieri, B. Montanari, F. Nocera, G. Sposito
yesterday and, in part, today, we completed the installation of the AdV+ PStab air-side cabling and EGO electronics.
There is a part of this job which will be done by the Artemis Team in the next few days.
We installed the PStab Interface unit on the INJ Tower rack, run the cables from there to both Laser Lab and EE Room (Servo and DAQ signals respectively), from the DAQ Room to the INJ Tower (DC Power for the in-vacuum electronics) and connected only on the Interface unit side the new one linking the in-vacuum portion of the system to the PStab Interface on the INJ Tower Flange and the ones going to the EE Room and Laser Lab (details below).
The test on the Interface unit has been only partial since there was no beam reaching the sensors on the suspended IB.
As requested by the Artemis Team, nothing has been removed so far.
Details on cabling:
to EE Room
- snake cable 8 STP labelled SDC004a.001
- snake cable 8 STP labelled SDC004a.002
to Laser Lab 4 STP labelled SDC003a.001