At the start of the shift I checked the alignment of the green (no manual alignment was needed) and then I locked the ITF in LOCKED_ARMS_IR_ALS.
Then a series of earthquakes in the Alaska region unlocked the ITF, the strongest one (magnitude 8.2, see attached picture) opened many SBE loops, the NE IP, the NE F7; to avoid further problems I manually opened all the SBE loops and also the NE_LC.
The effect of the earthquake was present until 11:30 UTC (see attached plot 2) and prevented any kind of ISC activities.
The SBE loops have been closed around 10:30 UTC; the SBE SNEB loop was closed thanks to help of Matteo because the horizontal loop was oscillating (SBE expert have been informed).
I fully recovered the NE suspension at 11:40 UTC; I realigned the cavities on the IR and I locked the the arms; I checked again the alignment of the green laser and finally at 12:03 UTC I locked the ITF in LOCKED_ARMS_IR_ALS.
From that point started the planned ISC activity.
Other parallel activities were on SQB1 which was vented and openend this morning.
High temperature in DER; experts verified that it is due to the presence of person in the lab.