bulten - 19:51 Wednesday 13 January 2021 (50435)
Print this report
SBD2 microsprings adjusted
after the recent works on SDB2, the controls went out of range; the thermal gradient on the GAS springs was too high. I adjusted the microsprings and left the controls in tracking on, loop closed. Actuator voltages are now small <0.5 V.